Product Specification Dimensions VisiLine series – Innovative functionality / flexible installation 26 12 3,5 ▪▪ Flexible assembly ▪▪ RGB and YUV interpolation algorithms on board ▪▪ Reliable transmission at 5000 Mbit/sec according to USB 3.0 standard ▪▪ Single cable solution for data and power 2 - M2 depth 3 Size Full Frames Resolution 4 - M3 depth 3 [max. fps] 3,5 8,7 1/4" 656 x 490 160 VLU-12M / VLU12C 1/3" 1288 x 960 42 640 x 480 376 1/3" 6,75 CMOS Sensor (monochrome / color) VLU-03M / VLU-03C 33 26 8 - M3 depth 3 C-Mount 26 VLU-02M / VLU-02C 11 CCD Sensor (monochrome / color) 18 14,2 Quick Start Guide VisiLine Cameras (USB3 Vision™) 33 26 Sensor Camera Type 48,3 9,8 Pixel 0,0 3,5 26 Latest software version and technical documentation available at: www.baumer.com/vision/login Safety System Requirements Installation Lens mount Safety Precautions Notice See the User's Guide for the complete safety instructions! CE We declare, under our sole responsibility, that the previously described Baumer VLU cameras conform with the directives of the CE. RoHS All VLU cameras comply with the recommendation of the European Union concerning RoHS Rules. CPU Single-camera system Multi-camera system Recommended Recommended Intel(R) Core(TM) i5-2520M Intel(R) Core(TM) i7-3770 CPU CPU @ 2.50GHz, Cores: 4 @ 3.40GHz, Cores: 8 4 GB 8 GB Caution RAM Observe precautions for handling electrostatically sensitive devices! Operating Microsoft Windows 7 32 / 64 bit systems (required for USB 3.0) system (OS) Microsoft® Windows® 8 32 / 64 bit systems (required for USB 3.0) ® ▪▪ Protect the sensor from dirt and moisture. ▪▪ Do not allow the camera to become contaminated with foreign objects. Environmental Requirements Storage temp. -10°C ... +70°C Operating temp. see Heat Transmission Humidity 10 % ... 90 % Non-condensing Further Information For further information about our products, please visit www.baumer.com For technical issues, please contact our technical support: email@example.com · Phone +49 (0)3528 4386-0 · Fax +49 (0)3528 4386-86 © Baumer Optronic GmbH · Badstrasse 30 · DE-01454 Radeberg, Germany Technical data has been fully checked, but accuracy of printed matter is not guaranteed. Subject to change without notice. Printed in Germany 01/15. v1.1 11135444 Notice Further technical details are available on the respective data sheets. ® Notice Ensure the sensor and lens are not contaminated with dust and airborne particles when mounting the support or the lens to the device! The following points are very important: ▪▪ Install the camera in an environment that is as dust free as possible! ▪▪ Keep the dust cover (bag) on the camera for as long as possible! ▪▪ Hold the printer with the sensor downwards if the sensor is uncovered. ▪▪ Avoid contact with any of the camera‘s optical surfaces! General Description LED signals 1 Heat Transmission Caution 2 Heat can damage the camera. Heat must be dissipated adequately to ensure that the temperatures do not exceed the values in the table below. As there numerous options for installation, Baumer does not specify a specific method for proper heat dissipation. As there are numerous possibilities for installation, Baumer do not specifiy a specific method for proper heat dissipation, but suggest the following principles: LED 3 ▪▪ operate the cameras only in mounted condition ▪▪ mounting in combination with forced convection may provide proper heat dissipation T 4 No. Description No. 1 Lens mount (C-Mount) 2 LED 3 Digial IO 4 Description Signal Meaning USB 3.0 port green USB 3.0 connection yellow USB 2.0 connection (settings possible, no image) LED Notice Why can frames not be transferred over an USB 2.0 connection? The camera needs to be supplied with more than 2.5W when transferring images. With an USB 2.0 connection maximally 2.5W are available. Therefore switching off of the frame transfer is necessary. However, adjustments are still possible. Installation Data Interface / Digital IOs Measurement Point Maximum Temperature T 50°C (122°F) Troubleshooting USB 3.0 Micro B 1 23 45 1 6 7 8 9 10 6 VBUS MicB_SSTX- Installing the camera: ▪▪ Connect the camera to the USB connection on your PC using an appropriate cable. → If LED is yellow: • Camera is connected to USB 2.0 (settings possible, no image). ▪▪ If required, connect a trigger and / or flash to the digital IOs. → If LED is green: • Check if camera is being used by another application. • Otherwise reconnect camera / restart software. 2 D- 7 MicB_SSTX+ 3 D+ 8 GND_DRAIN 4 ID 9 MicB_SSRX- 5 GND 10 MicB_SSRX+ Digital IOs (M8 / 8 pins / wire colors of the connecting cable) 5 4 2 7 1 8 1 OUT 3 5 IO Power VCC grey 2 not connected brown 6 OUT 1 pink 3 IN 1 green 7 not connected blue 4 IO GND yellow 8 OUT 2 red white Installation example 1 - PCI USB board 2 - USB cable 3 - Cable for trigger and flash 3 6 1. Check camera operation using the LED signals. open wire 2. Check connection using Windows Device Manager: → If device is not listed: • Check the host controller power supply. • Check USB 3.0 cable and connection. → If device is regularly not listed • Check USB 3.0 driver installation.