ASM-SENSOR 3284

```3284 [200A rms, ø 33mm max.]
3285/-20 [2000A rms, ø 55mm
(1.30")
(2.17") max.]
CLAMP ON AC/DC HiTESTER
2000 A analysis
Field measuring instruments
(3285 /3285-20)
3284
3285-20
Analysis for
CAT III 600 V
DC to distorted
waves
http://www.hioki.co.jp/
3285
����������������������������������������������������������������������
������������������������������������������������
Waveform harmonic
Analysis applications
and
Supporting a wealth of functions
Inrush current
0mA
The peak hold function displays
the crest value of the inrush
current occurring when electrical
equipment is started.
8807-51
MEMORY HiCORDER
crest value
9094
OUTPUT CORD
+
Crest
value
peak
Using the external output functions of the 3284 or 3285 in combination
with a HIOKI MEMORY HiCORDER enables recording of current
and frequency fluctuations and recording and harmonic analysis of
instantaneous waveforms. The monitor output (MON) spans a wide
frequency range from DC to 20 kHz (15 kHz in the case of 3285), which
allows the units to be used for electric current analysis in various fields
ranging from batteries to inverters.
0
+
0
Example of analysis using the MEMORY HiCORDER
RMS value of
full-wave rectified waveforms
current waveform
-
The AC+DC mode enables
measurement of the RMS value
of full- or half-wave rectified
waveforms used in electrical
machinery.
(inverter output )
▼
+
▲
current fluctuations
0
■ 3284 / 3285 Specifications
Function Mode Display Switch*1
Voltage
(V)
Frequency (Hz)
Resistance(Ω)
*8
Continuity Check*8
DC
Average / PEAK
AC
RMS / PEAK
AC+DC RMS / PEAK
[ ] in the Specification :3285,3285-20
Accuracy at 23 ˚C ± 5 ˚C (73 ˚F ± 9 ˚F), 80% rh or less.
Measurement accuracy applies to input of at least 5% of the current range, and at least 10% of the voltage/frequency range.
Range
Accuracy at DC or 45 to 66 Hz
20.00 / 200.0
[200.0 / 2000]
±1.3%rdg.±3dgt.
Resolution is
0.1 [1] A during PEAK
±1.3%rdg.±7dgt.
Max. allowable current
±1.3%rdg.±3dgt.*2
±2.0%rdg.±5dgt.*3 *4
±1.3%rdg.±7dgt.*6
200Arms cont. 300A peak max.
±1.3%rdg.±13dgt.*2
±2.0%rdg.±7dgt.*3*4
±1.3%rdg.±7dgt.*6
[2000Arms cont. 2840A peak max.]
±1.0%rdg.±7dgt.
Max. allowable voltage
600Vrms cont. 1000V peak max.
*6
Average / PEAK 30.00 / 300.0 / 600
±1.0%rdg.±3dgt.
AC
Resolution is decreased by
1 digit except in the 600 V
range during PEAK
±1.0%rdg.±3dgt.
±1.5%rdg.±5dgt.
±1.0%rdg.±7dgt.
±1.0%rdg.±7dgt.
±1.5%rdg.±7dgt.
±1.0%rdg.±7dgt.
10.00 / 100.0/1000
±0.3%rdg.±1dgt.
AC+DC RMS / PEAK
Notes
10 to 45Hz or 66 to 2 [ 1 ] kHz PEAK accuracy at DC or 45 to 66 Hz
DC
RMS / PEAK
▲
Harmonic analysis
-
Current
(A)
analysis
Model 3284 and 3285 only
9199
BNC TO BANANA
*5
*5
1000/10.00k
1000
*7
±1.0%rdg.±1dgt. (at 1000Hz)
±1.5%rdg.±5dgt.
Threshold level 30Ω ±5Ω or less
*7
Effective in the V and A functions
Open terminal Voltage 3.1V DC or less
*1 The peak hold function works automatically during PEAK measurement. *2 Add ±1.0% rdg. in the 1800 to 2000 A interval for the 3285. *3 Add ±1.0% rdg. in the 1k to 2kHz interval
for the 3284. *4 No assured accuracy in the 1800 to 2000 A interval for the 3285. *5 Add 6 dgt. in the 30 V range. *6 Add ±4.7% rdg. in the 2300 to 2840 A interval for the 3285. *7 Add ±
1.0%rdg.±13dgt when measuring 700 V to 1000V with Model 3285-20. *8 Model 3285-20 only.
Measurable conductor diameter : ø33 (1.30") mm, [ø55 (2.17") mm] max.
Max. usable circuit voltage : 600Vrms (insulated conductor)
Functions : recording (MAX. MIN. AVE) ,data hold, auto power-off, auto zero
Display update rate : digital indication FAST approx. 4 times/second, NORMAL approx. 2 times/second,
SLOW 1 time/ 3 seconds; bar graph indication approx. 4 times/second
Display response time : 1 sec or less
Range switching : Auto/Manual (Except resistance.)
Frequency characteristic : 40 Hz to 2 [1] kHz accuracy
External output : A/Hz only: recording output DC1V/f.s. ±1.3% rdg.±10mV, Response: 250ms
(Except model 3285-20) or less (at 0➞90%)/ MON A only : Waveform output 1Vrms/f.s. ±1.3% rdg.±5mV,
Frequency band: DC to 20 [15] kHz ±3dB
3284
CLAMP ON AC/DC HiTESTER
3285
CLAMP ON AC/DC HiTESTER
3285-20 CLAMP ON AC/DC HiTESTER
(All include 9207-10 TEST LEADS, 9399 CARRYING CASE(3284), 9345 CARRYING CASE(3285/3285-20), HAND STRAP)
Effect of conductor position : within ±0.5[0.7] % in any direction from the center of sensor
Effect of external magnetic field : 400A/m corresponds to 0.5 [2] A max.
Crest factor : 2.5 or less, 1.7 or less in the 600 V range, 3284 is 1.5 or less in the 200A range [1.42 or less in the 2000A range]
Safety standard : EN61010-1 CAT III 600V, EN61010-2-032, EN60529 IP40
EMC : EN61326, EN61000-3-2, EN61000-3-3
Ambient temperature of use : 0 to 40˚C (32˚F to 104˚F), 80%rh or less (no condensation)
Storage temperature range : -10 to 50˚C (14˚F to 122˚F) or less (no condensation)
Power supply : 9445 AC ADAPTER(Except Model 3285-20.) or 6F22 manganese battery×1
(continuous operation : Model 3284/3285-25 hours, Model 3285-20-20 hours)
Dimensions and mass : Approx. 62W×230[260]H×39D mm, 460 [540]g
(Approx 2.45" W× 9.06 [10.3]" H× 1.54" D, 16.3 [19.1] oz.)
Options
An AC Adapter can only be used with Models 3284 and 3285.
*
9094 OUTPUT CORD(Jack to banana plug)
*No CE marking
DISTRIBUTED BY
81 Koizumi, Ueda, Nagano, 386-1192, Japan
TEL +81-268-28-0562 / FAX +81-268-28-0568
E-mail: os-com@ hioki.co.jp
HIOKI USA CORPORATION :
6 Corporate Drive, Cranbury, NJ 08512 USA
TEL +1-609-409-9109 / FAX +1-609-409-9108
E-mail: hioki@ hiokiusa.com
Shanghai Representative Office :
1704 Shanghai Times Square Office