3286-20 CLAMP ON POWER HiTESTER Power measuring instruments All Powerful Effective input current level 1.00 A to 1000 A W VA var 600.0kW max.(1ø) COS ±1.000 ±1.000 (±90.0º) (±90.0º) Harmonics Up Up to to 20th C A T III 6 0 0 V http://www.hioki.co.jp/ 1 Functionality and Safety 80 mm ✕ 20 mm busbar max. Applications ø 55 mm max. ❏ Use as a single-phase power meter RMS V Effective power/ voltage/ current Load AC 1ø Power meter mode Source Voltage 600 Vrms max. Current AC RMS A 1000 Arms max. This displays the result of vector calculations on PT/CT conversion ratio settings are not possible. single-phase power. Apparent/ reactive power Power factor (effective value power factor) Power W VA var 600.0kW max.(1ø) Phase detection ❏ Measure phase on a single-phase line RST 1ø Power Factor meter mode The phase angle between the voltage waveform and the current waveform is found, and the power factor (cos ø) and reactivity (sin ø) are calculated and displayed. Normal/ Reverse/ Missing Power factor (cos ø) Power factor COS ø ±1.000 (±90.0º) Peak Peak Reactivity (sin ø) Phase angle (ø) value 1700 A peak max. Frequency Hz 1000 Hz max. Harmonics Harmonics Up to 20th The 3286-20 provides the following 2 types of power factor measurement. If the waveform includes harmonic distortion, there will be a difference between the two measurement values, and the method must be selected accordingly. ●Power factor derived by phase discrimination (cos ø): 1ø Power Factor meter and 3ø Power Factor meter mode From the phase information for the fundamental frequency component, cos ø is calculated. This is suitable for assessing the quality of a power supply. ●Power factor (effective value power factor) derived from effective power ÷ apparent power: 1ø Power meter mode This is calculated from the power value including harmonic components (effective power ÷ apparent power). This is suitable for evaluating the performance of standalone load equipment and similar cases. * It is not possible to calculate the effective power factor of a three-phase line. 2 Packed into a Handheld Unit Helpful battery capacity check Whenever the unit is powered on, and while the record function is operating, the battery capacity can be checked to avoid battery exhaustion during measurement. ❏ Simple checking of three-phase lines ❏ For harmonic suppression 3ø Power Factor meter mode Load Source Power factor (cos ø) Harmonic measurement function Harmonics effective value/ total harmonic distortion Harmonics effective value/ harmonic percentage Phase angle (ø) Reactivity (sin ø) Harmonic coefficients Total harmonic distortion Ex. THD-R is 65.9% Ex. Fundamental Harmonic coefficients Harmonic percentage Ex. 3rd order is 33.7 A component is 40.3 A Balanced three-phase power can also be displayed Effective power/ voltage/ current Apparent/ reactive power Phase detection Ratio to case in which fundamental component is 100% The analysis values for coefficients for each of the harmonics from 1 to 20 of voltage or current can be displayed. ❏ External output of data Output to PC (Optional 9636-01 required) check The 3286-20's three-phase power measurement method calculates and displays the power values for a sine wave input at 50/60 Hz, assuming it is balanced and there is no distortion. Accurate measurement is not possible on a three-phase line if it is not balanced, for example when controlled by an inverter or thyristor. Cord length: approx 2m Since there is no integration function, it is not possible to measure total energy consumed (Wh). ❏ Check power supply fluctuations Max. and min. value displays Using the record function, it is possible to check the amplitude of fluctuations. Record elapsed time Example: 4 h 25 m V Waveform display is not available Care must be taken with regard to battery life Record peak value fluctuations Easily check peak value fluctuations. Please note that waveforms with amplitudes at or less than 250ms cannot be detected accurately + Data output connector RS-232C interface by optical insulating coupler ● OS:Windows 95, 98, NT4.0, 2000, Me*1 ● Connector: D-sub 9 pin ● Fetch interval: 6s/30s/1m/5m/10m. ● Buffer size: 32,700 max. Care must be taken with regard to battery life *1. Windows is a registered trademark of Microsoft Corporation. Printer output to 9442 (optional) The output is sent to the printer each time the 3286-20 HOLD key is pressed. (For about 1 sec.) ● Printing method: Thermal serial dot matrix ● Paper width: 112 mm ● Printing speed: 52.5cps ● Power supply: 9443 AC ADAPTER or supplied nickelhydride battery (capable of printing about 3000 lines on full charge from 9443) ● Dimensions and mass: 160W ✕ 66.5H ✕ 170D mm; 580 g 9636 9443-02 RS-232C CABLE AC ADAPTER (For the EU) 9443-03 (For the America) 0 - Waveform display is not available Cord length approx 1.3m Not for use with Model 3286-20 3 ■ Basic specifications Measurement items : Voltage, current, voltage/current peak, effective/ reactive / apparent power(Single-phase or 3-phase), power factor, reactivity, phase angle, frequency, phase detection(3phase), voltage/current harmonic levels(up to 20th) Measurable conductor diameter : ø55mm (2.16"), 80 mm ✕ 20 mm busbar max. : LCD, digital (6000 counts) Display Rectification method : RMS (true root mean square value) Display update rate : NORMAL approx. 1 time/ sec, SLOW 1 time/ 3-sec at HARM meas. approx. 1 time/ 2-sec Analog response time : 4.0 seconds or less (when input is changed from 0% to 90% of range.) [Voltage/ Current/ Power measurement] AC Voltage Range Table 150.0 V Detection method : Power factor (cos ø) : : Phase angle Reactivity (sin ø) : Phase discrimination by phase detection (zero crossing) 0.000 (lead) to 1.000 to 0.000 (lag) 90.0˚(lead) to 0.0˚to 90.0˚(lag) 0.000 (lead) to 1.000 to 0.000 (lag) [Frequency measurement] Effective in the voltage and current functions Measurement range : 30.0 Hz to 100 Hz (at 100.0 Hz range) 100 Hz to 1000Hz (at 1000 Hz range) Min. input level : Voltage 10 Vrms-sine wave, Current 1 Arms-sine wave [Wave peak measurement] Effective in the voltage and current functions AC Current 20.00 A 200.0 A 1000 A Single-phase 3.000 kW *3-phase (balanced load) 6.000 kW 30.00 kW 150.0 kW 60.00 kW 300.0 kW 300.0 Single phase 6.000 kW 60.00 kW 300.0 kW V *3-phase (balanced load) 6.000/12.00 kW 60.00/120.0 kW 600.0 kW 600 V Single phase 12.00 kW *3-phase (balanced load) 24.00 kW [Power factor/ Phase angle/ Reactivity measurement] 120.0 kW 600.0 kW 240.0 kW 600.0/1200 kW *3-phase power is calculated and displayed on the basis of a balanced, 50/60 Hz, sine wave input. For apparent power and reactive power, the unit of watts in the above table is replaced by VA and var respectively. Effective value P.F. : Max. allowable current : Max. usable circuit voltage : Effective input range : 0.000 (lead) to 1.000 to 0.000 (lag); 1ø only 1000 Arms cont. 600 Vrms (insulated conductor) Voltage: 10 V to 600 V, Current: 1 A to 1000A, Power: 80 V to 600 V and 1 A to 1000 A Min. Display value : Voltage: 0.6 Vrms, Current: 0.06 Arms Display indication range : 5 or less are zero-suppressed, and the upper limit is to 125% of the range setting (to 100% for the 1000 A range) (RMS value) Circuit dynamic : 2.5 or less (1000 A and 600 V range is 1.7 or less) Measurement range : 150 (375 peak) / 300 (750 peak) / 600 (1020 peak) V 20 (50 peak) / 200 (500 peak) / 1000 (1700 peak) A Effective Input Range : Effective value of sine wave is within effective input permissible in the range and within circuit dynamic [Harmonic measurement] Effective in the voltage and current functions Measurement items : Level of each order, percentage of each order and total harmonic distortion (THD-F and THD-R) Measurement range : Fundamental frequency 50 / 60 Hz Window width : 1 cycle (50 / 60 Hz), Data points: 256 points Window type : Rectangular Orders analyzed : Up to 20th [Other functions] Phase detection : Normal/ reverse/ missing (at 3-phase balanced load) : MAX. value and MIN. value (Effective in the voltage, Record current and effective / apparent power functions) Battery capacity : Displayed in % when the unit is powered on : Holds display Data hold Auto power off : Approx. 10 minutes, buzzer sounds just before power is turned off, can be extended or cancelled : RS-232C interface by optical insulating coupler Data output ■ Measurement accuracy (23 ˚C±5 ˚C (73˚F ±9˚F), Less than 80%rh., sine wave input, power factor = 1) 30 Hz to 45 Hz 45 Hz to 66 Hz 66 Hz to 1 kHz Voltage ±1.5 %rdg.±5 dgt. ±1.0 %rdg.±3 dgt. Current not rated ±1.5%rdg.±5 dgt. ±1.3 %rdg.±3 dgt. ±2.0 %rdg.±5 dgt. Accuracy guaranteed only for 50/60 Hz (COS ø=1) Power : ±0.3 % rdg.±1 dgt. (at 100.0Hz range) ±1.0 % rdg.±1 dgt. (at 1000Hz range) : ±3.0 %rdg.±5 dgt. (45 Hz to 1 kHz) Wave peak Thermal coefficient : Voltage and current: within ±0.1✕Accuracy/ ˚C (0 to 40˚C) Phase: within ±2˚ (0 to 40˚C) Conductor position : Within ±0.7 % in any direction from the center of sensor External magneticfield : 400A/m corresponds to 1.00 A max. Power factor influence : ±10.0 % f.s. (cos ø=0.5) : Order Harmonics Order Accuracy Accuracy Frequency [Voltage/ Current/ Power measurement] Single-phase: ±2.3 %rdg.±5 dgt. 3-phase: ±3.0 %rdg.±10 dgt.(at balanced load) 1 : ±3˚ Phase angle Power factor (cos ø) : ±3˚±2 dgt. 2 to 6 7, 8 ■ General Specifications ±3.0 %rdg.±10 dgt. ±3.5 %rdg.±10 dgt. ±4.5 %rdg.±10 dgt. 9, 10 11 to 15 16 to 20 ±5.0 %rdg.±10 dgt. ±7.0 %rdg.±10 dgt. ±10.0 %rdg.±10 dgt. Applicable standards : Safety EN61010-1:1992+A2:1995 CAT III 600V EN61010-2-031:1996, EN61010-2-032:1995 EN60529:1991 IP40 : EMC EN61326-1:1997+A1:1998+A2:2001 Withstand voltage : 5.55 kV AC between clamp and frame, between (50/60 Hz, 1 minute) clamp and circuitry Measurement method : Digital sampling method Operating temperature : 0ºC to 40ºC, 80%rh or less, no condensation Storage temperature : -10ºC to 50ºC, no condensation Power supply : 6LR61 alkaline battery✕1 (continuous operation max. 25 hours) (9V battery) 6F22 manganese battery✕1 (continuous operation max. 10 hours) Dimensions, mass : Approx 100 W ✕ 287 H ✕ 39 D mm, Approx 650g 3286-20 CLAMP ON POWER HiTESTER Options 9635-01 VOLTAGE CORD 9636-01 RS-232C PACKAGE 9442 PRINTER (Includes 9635 VOLTAGE CORD, 9245 CARRYING CASE, HAND STRAP) (Approx 3.94″(W )11.3″ (H)1.54″ (D), Approx 22.6 oz.) 9636 9443-02 9443-03 1196 Cord length approx 3m 9635-01 RS-232C CABLE (For 9442 printer) AC ADAPTER (For 9442 printer, EU) AC ADAPTER (For 9442 printer, America) RECORDING PAPER (For printer, 10 rolls) Cord length approx 3m When ordering the 9442 PRINTER, also order the 9636 RS-232C CABLE required for connection to the 3286-20, and 9443 AC ADAPTER. DISTRIBUTED BY HEAD OFFICE : 81 Koizumi, Ueda, Nagano, 386-1192, Japan TEL +81-268-28-0562 / FAX +81-268-28-0568 E-mail: [email protected] hioki.co.jp HIOKI USA CORPORATION : 6 Corporate Drive, Cranbury, NJ 08512 USA TEL +1-609-409-9109 / FAX +1-609-409-9108 E-mail: [email protected] hiokiusa.com Shanghai Representative Office : 1704 Shanghai Times Square Office 93 Huaihai Zhong Road Shanghai, 200021, P.R.China TEL +86-21-6391-0090, 0092 FAX +86-21-6391-0360 E-mail: [email protected] All information correct as of Oct. 3, 2003. All specifications are subject to change without notice. 328620E2-3XB-05K Printed in Japan

- Similar pages
- CLAMP ON POWER LOGGER PW3360-20
- ASM-SENSOR 3169-20
- ASM-SENSOR 8807-51
- ASM-SENSOR 3390
- ASM-SENSOR 3197
- ASM-SENSOR 3193
- ASM-SENSOR 3196
- ASM-SENSOR 3334
- ASM-SENSOR 8714-01
- ASM-SENSOR 3282
- ASM-SENSOR 3283
- ASM-SENSOR 3284
- JDSU SPM-33A
- ASM-SENSOR 3511-50
- ASM-SENSOR 3156
- ASM-SENSOR 3560
- ASM-SENSOR 3532-50
- pdf-download 3286-20
- ASM-SENSOR 3174
- ASM-SENSOR 3159
- ASM-SENSOR 9334
- DIGITAL M-ohm HiTESTER 3154